FAQ’s to help with trouble shooting and self service:
How to reload the DTM's calibration factors http://www.group3technology.com/vdb/document/64
Reset your DTM by placing SW2/8 to the on position and power cycle.
Check grounding or any other source of interference.
Ensure your DTM is as far away as possible from the magnetic source.
Download and read the "BADCAL Error Correction Proceedure
Your Hall Probe Temperature sensor has failed and requires repair.
Your DTM may have lost its calibration or requires a reset please see Q1 and Q3 above.
Please contact your local dealer or distributor to arrange a service.
Copyright © 2015 - 2018 Group 3 Technologies. All rights reserved | Sitemap | Privacy | Website Design by MOCA